Some compounds absorb light with wavelengths in the ultraviolet and/or visible range of the electromagnetic spectrum (200nm to 780nm).
UV-Visible spectroscopy uses a UV-visible spectrometer to give a measurement of the sample's absorption.
The amount of light absorbed is proportional to the concentration of the compound in solution.
Concentrations are often expressed as mg/L or ppm.
The amount of light absorbed by the sample is compared to the amount of light absorbed by a set of standards of known concentration.
Toluene is an important industrial chemical.
The concentration of toluene in a solution can be determined using UV-visible spectroscopy.
Choose an appropriate wavelength: a wavelength at which there is significant absorbance .
The UV-visible spectrum of a solution of 7.5 ppm toluene is shown on the right.
It can be seen that the maximum absorbance of toluene occurs at 265 nm.
265 nm is the best wavelength to choose in order to determine the concentration of toluene in solution.
Prepare a set of standard solutions of toluene.
A standard solution is a solution with a known concentration.
Measure the absorbance of each standard solution at the selected wavelength (265 nm).
It is usually best to measure the absorbance of each solution several times and then take the average absorbance to reduce errors.
Plot the calibration graph.
Measure the absorbance, at the same wavelength, of the sample solution of unknown concentration.
At 265 nm the solution of unknown toluene concentration had an absorbance of 1.28
Plot the absorbance of the sample solution on the calibration graph and read off the concentration.
The concentration of toluene in the sample solution is 12.0 ppm.